Medical device CE/IVD marked
- Product Number: 227-40110
- Component of DOSIMMUNE™ reagent kit
DOSIMMUNE™ is registered as a CE/IVD device in the EEA. In other case, DOSIMMUNE™ is intendent for research use only, not for diagnostic.
DOSIMMUNE™ Calibrator Set is to quantify immunosuppressants (Cyclosporin A, Everolimus, Tacrolimus, Sirolimus) in human whole blood by LC-MS/MS method.
The set is composed by 6 calibrators with different concentration levels and 1 blank level.
The kit contains all material needed for quantifying the four immunosuppressant drugs.
Product available for CE/IVD 227-40110-55 or RUO 227-40110-58 (Europe) / 227-40110-91 (Japan)
Content: 6 calibrators level (lyoph.) 6x500 µL and 1 blank (lyoph.) 500 µL
Parameter: Cyclosporin A, Everolimus, Tacrolimus, Sirolimus
|Parameters||Blank||Level 1||Level 2||Level 3||Level 4||Level 5||Level 6|
|Cyclosporin A||/||25 µg/L||100 µg/L||450 µg/L||1000 µg/L||1500 µg/L||1800 µg/L|
|Everolimus||/||2 µg/L||5 µg/L||10 µg/L||15 µg/L||25 µg/L||35 µg/L|
|Tacrolimus||/||2 µg/L||5 µg/L||10 µg/L||15 µg/L||25 µg/L||35 µg/L|
|Sirolimus||/||2 µg/L||5 µg/L||10 µg/L||15 µg/L||25 µg/L||35 µg/L|
Reconstitution: Remove carefully the cap of each vial and reconstitute each calibrator with exactly 500 µL of ultra-pure water. Shake during at least 20 minutes at ambient temperature until the content becomes homogeneous.
Storage: refer to the Instruction for use
Comment: The Calibrator Set DOSIMMUNE™ is delivered with a safety data sheets to use properly the product.